|Display Device:||Color LCD Display + Touch Screen||Temperature Control Range:||RT-150℃|
|Temperature Control Accuracy:||±0.02℃||Test Hole Number:||4 Hole|
|Time Accuracy:||0.1 Seconds||Storage Capacity:||2000 Test Records|
|Power Supply:||AC220V 5% 50Hz 5%||Power:||1500W|
|Weight:||About 50kg||Print Mode:||Panel Type Thermal Printer|
abrasion testing machine,
adhesion testing equipment
220V Kinematic Viscometer For Liquid Material With Temperature RT-150℃ And High Precision For 0.02
Kinematic Viscometer for Petroleum Products is suitable to national standard GB/T265-1988
1. Applied to the determination of the viscosity of liquid petroleum products (Newton liquid). The instrument adopts fuzzy PID control technology, advanced precise temperature control, can also be used as a common thermostatic water bath.
2. The instrument can be enabled at the same time 4 pieces of glass capillary viscometer was under the same temperature determination of kinematic viscosity and viscometer can be repeatedly measured 4 times as a set of test data. The instrument adopts high definition color touch screen technology, the interface is rich, clear, touch screen response is fast, flexible, and the use of large capacity storage chip, can save 2000 of the results, the use of the panel mounted thermal printer, the end of the test, automatic rapid printing test results.
|Display device||color LCD display + touch screen|
|Temperature Control Range||RT-150℃|
|Temperature Control Accuracy||±0.02℃|
|Test hole number||4 hole|
|Time range||0~999 seconds|
|Time accuracy||0.1 seconds|
|Storage capacity||2000 test records|
|Print mode||panel type thermal printer|
|Power Supply||AC220V 5% 50Hz 5%|
|automatic rapid printing test results|
Obverse chart (chart 1)
1. Roof assembly 2. Temperature sensor 3. Water bath 4.Printer 5. Touch liquid crystal display 6. Power switch
Reverse chart (chart 2)
1. Cooling fan 2. Power switch 3. Power supply socket 4.Fuse tube 5,6. Control cable socket
(3) Water bath cover chart(chart 3)
1. control box 2. Temperature sensor jack 3. No. 4 hole viscometer 4. No. 3 hole viscometer 5. Control cable socket 6. Mercury thermometer jack 7. No. 1 hole viscometer 8. No. 2 hole viscometer
After the start, click on the screen at any position, enter the functional selection interface to select the menu item. The interface has three menu items: Testing,Dead Work and Data Record.
Click the ‘Testing’ button in the functional selection interface to enter the test operation interface.
As shown above, there are 4 buttons in the interface, ‘T.control’ (refers to ‘temperature control’), ‘Stop’, ‘Para.’(refers to ‘parameters’), ‘Exit’.Click on ‘T.control’, set the temperature of the constant temperature, at this time, the ‘T.control’ button failure. If you need to stop temperature control,you can click on the ‘Stop’, at this point, the ‘stop’ button failure, ‘T.control’ for the effective state. Click on the ‘Para.’ to enter the 4 glass tube diameter, constant set interface. Click on ‘Exit’ to exit the interface. The upper part of the screen shows the real time temperature. Below the ‘temperature’ is the set value of temperature. Four timers side by side (number 1, 2, 3, and 4 from the left and right in turn corresponds to a water bath cap 4 pieces of glass viscometer) with a resolution of 0.1 of a second time, four timer current time values are just below timer dial. Click on the dial to start time, again click the time to stop, and the timer has been ejected.Such as no operation, after a few seconds the interface will automatically exit. I f the instrument hase reached the number of timing, the instrument will pop out the results interface of the calculation of the viscosity . At any time, click on the time just below the clock dial,you can get into the time interface as well.
Time interface is as follows:
In this interface ‘(1)’, ‘(2)’, ‘(3)’, ‘(4)’refers to the 4 consecutive time of the same viscometer. Click on the "Delete", delete the corresponding time (in that the timing is obviously wrong).
Kinematic viscosity calculation results interface shows the kinematic viscosity results of corresponding glass viscometer, the unit is mm2/s. Click on "exit" to return to the test interface. If you don't click on ‘Exit’, after a few seconds, automatically exit.
Click on the real time displayed in the middle of the test interface, enter the time setting interface:
Click on the real time at the top of the screen, the digital input area above the soft keyboard shows "20?? -?? -?? ??: ?? :??". In order to enter the number of years, months, days, hours, minutes, seconds, click on‘Enter’, the time is modified. Click on ‘Exit’ to exit the time setting interface.
(3) Prepare working interface:
The interface contains four functions: temperature set, printer set,
viscometer parameters (diameter and constant) setting and number of timing (with a glass capillary viscometer with a group of test taking multiple timing value).
constant temperature setting interface:
The symbol "√" is always located in the back of the set value .When you click on the right bottom of arbitrary temperature setting value, enter arbitrary temperature setting interface.
After entering the printer settings interface, select ‘Enable/Disable printer". Test end, the instrument print(The printer is set to’Enable’) or not print(The printer is set to ‘Disable’) measurement results.
Viscometer parameter settings:
The left column is the glass capillary viscometer diameter, unit: mm. The right column is constant of the corresponding viscometer, unit: mm2/s2. Click the parameter to be modified, is transferred to the rear of the parameter, and the pop-up interface of the parameter changes is displayed. In the interface to modify the parameters, first of all click on the left side of the parameter display area, then you can enter the corresponding digital, and click the "←" to confirm. To cancel the input, click the "CE" button.
d. Setting of timing number:
The timing number refers to consecutive timing number of the same glass viscometer, range: 1~4. There are two kinds of methods: a. direct calculation of kinematic viscosity average;b. Calculating the average value of the kinematic viscosity after abandoning the super difference of the time.
(4) Data record interface:
The interface shows the records of each test, including the test serial number, the test time and the kinematic viscosity value. There are 4 functional buttons in the interface:’Pg Up’(page up),’Pg Dn’(page down),’Clear’and ‘Exit’----- Click on the ‘ Pg Up ‘,the data record page up; Click on the ‘Pg Dn’, the data record page down;Click on the ‘clear’ ,the remind dialog box of deleting record pops up. In the dialog box, click on the ‘Enter’, and the data will be cleared; Click on the ‘Cancel’ to exit the dialog box; Click on the ‘Exit’ to exit the data record interface.
About our factory:
Dongguan Liyi Environmental Technology Co., Ltd is a high technology supported production-oriented enterprise who specializes in developing, manufacturing,marketing all kinds of environmental equipment.Our company has advanced technology, experienced technical management personnel and efficient professional service staffs, fine quality management and perfect after-sales service system.
To strengthen the quality awareness,perfect management system,improve product credit,meet customers increasing requirements, our company brought into ISO9001:2008 international quality management system and obtained the certificate: N0006997-1.
Our company will continue to develop, innovate and keep faith. We combine with the international testing syandards and the specific demands of customers that may establish a new banner in Chinese testing industry.
What is the mini order quantity?
MOQ: generally 1 unit/piece/set
What payment methods are supported?
Payment method: By T/T, L/C, Alibaba Trade Assurance etc.
Payment terms: Generally, for test instruments T/T 100% advanced; or 100% irrevocable L/C at sight.
What are the available delivery methods?
Our port location: Chongqing, Shanghai or other main port of China
Delivering to: Worldwide
Delivering method: by sea, by air, by truck, by express, combined transportation
How long is the product date of delivery probably?
Delivery date differs with product type, order quantity, special requirement etc. Usually our big machine delivery date is around 10 ~ 20 days; test equipment or analyzer delivery date is around 5~10 days. Some products have stock, contact us anytime to get more information.
How long is the warranty period?
We warrants the plant supplied under agreed specification against defects in material and workmanship under normal use and service for a period of 2 years after start up of the system.
How is the technical services on the product?
Engineer available to service machinery overseas. If you have any technical question, you are welcome to contact us any time. We will response promptly and answer to your satisfaction. If required and necessary, we can send our engineer to your place to provide technical support.
Contact Person: Victoria Li