|Press Needle Diameter:||1.25mm||Press Needle Telos Diameter:||0.79mm±0.03mm|
|Press Needle Taper Angle:||35°±0.25°||Press Needle Stroke:||2.5mm±0.045|
|Dial Range:||0HA-100HA6||Net Weight:||0.5kg|
automatic hardness tester,
digital hardness tester
1.25mm Press Needle Hardness Testing Machine Shore For Rubber
This tester is used for testing the hardness of vulcanized rubber and plastic products.
1. The gauge outfit of the durometer is installed on the top of the rack.
2. It is more convenient to operate and test result more precision. Also can take down the gauge outfit and
test when production.
GB/T531, GB2411, ASTM D 2240-5, JISK 6253-2006, ISO 7619-1-2004, JJG 304-2003
|Press needle diameter||1.25mm±0.15mm|
|Press needle telos diameter||0.79mm±0.03mm|
|Press needle taper angle||35°±0.25°|
|Press needle stroke||2.5mm±0.045|
Method of measurement
This hardness tester like many other hardness tests, measures the depth of an indentation in the material created by a given force on a standardized pressure foot. This depth is dependent on the hardness of the material, its biscoelastic properties, the shape of the pressure foot and the duration of the test. Shore durometers allows for a measurement of the initial hardness or the indentation hardness after a given period of time. The basic test requires applying the force in a consistent manner,without shock measuring the hardness(depth of the indentation).If a timed hardness is desired, force is applied for the required time and then read.
Contact Person: Kayle Huang